This GRUNDIG UV5A Owner Manual is not only an instruction manual but a total functional circuit description of the Multimeter and includes circuit diagram at the end pages. It is very helpful for repairing and calibrating the instrument. It is written in two languages English and German for international support. It was very easy to repair my unit with this document. Regards, Regis Pauly, Electrical Engineer.
Fantastic quality and impressive delivery time. Based on this manual, I recommend this service without hesitation! Five Stars for certain.
Text excerpt from page 5 (click to view)
Connecting to electricity
The oven is designed to be connected to 240 V - 50 Hz electricity supply. The oven has an easily accessible terminal block which is market as follows. Letter L Live terminal Letter N Neutral terminal Earth The cable used (of the type H05 RR-F) must have a min. cross section of 1.5 mm2. In order to connect it to the oven it is necessary to remove the hatch at the back of the oven to reach the terminal block. For direct connection to the mains supply a two pole swich whith a minimum rating of 13 A and a minimum distance of 3 mm between its contacts must be fitted between the supply and appliance. This swich must not break the yellow/green earth cable at any point. The manufacturer disclaims any responsibility should these safety instructions not be carried out.
The appliance must be earthed
Important After installation and connecting, the cable must be placed so that is cannot at any point reach a temperature more than 50Â°C above the ambient temperature. Before the appliance is connected check that: the main fuse and the domestic installation can support the load; the power supply is properly earthed; the double pole swich used to connect the appliance to the supply is easily accessible after installation.